Send comments to H. Huang,
1. Open GPM of BeamDecay to monitor the impact on beam from target insertion. Make sure the beam decay goes away after the measurment. If not, the pet page should be used to rotate target out( from Summary tab, click Target, then find the active target, select rotateout).
When changing values in a field, the field turns yellow (ala pet). Hit the return key in that field to "activate" that value. If the cell remains with a yellow background after pressing the return key, the value entered is most likely not valid and a different value should be used.
When selecting the target from the target scan page doesn't seem to "take", try selecting the other orientation (Horizontal/Vertical) tab and then go back to the original tab and try selecting the target again.
There are 3 servers involved when using the polarControl program.